AD frequency range: MHz. Square wave: MHz. AD adopts MHz Active Crystal. The module can output two sine waves and two square. IC AD is an ADI DDS chip which be common used in many DDS projects. This AD DDS module is basic on this ADI AD chip. This module provide. IC AD is an ADI DDS chip which be common used in many DDS projects. This AD DDS module is basic on this ADI AD chip.
|Genre:||Health and Food|
|Published (Last):||9 December 2004|
|PDF File Size:||9.90 Mb|
|ePub File Size:||13.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
For more information regarding pin functionality please consult the datasheet linked to above. Some failed to produce any output at all. Affordable portable and bench scopes–DIY kits available! The important modulle are broken out to 0. Please, log in or register.
Write Your Own Review Only registered customers who has bought this product can write reviews. I accept the Terms and Coditions. Since it may not be easy for many kit constructors to carry out this adjustment, and there are already enough variables, it seems only polite for QRP Labs to supply the module ready-adjusted.
AD9850 0-40MHz DDS Module
The adjustment is almost correct here, but still slightly too clockwise. App Notes Discussion group Shipping information Beware imitations!
For technical support, please open a ticket on Itead Support System. Only registered customers who has bought this product can write reviews.
Turn the potentiometer anti-clockwise “left” carefully, until a pulse waveform is seen. In practice, seconds is sufficient to show any instabilities. It will be appreciated if you can help us improve the xds by correcting the errors, adding more demo codes or tutorials. The second dit speed ensures that nothing could possibly change state that could interfere with any measurement. Write Your Own Review Only registered users can af9850 reviews.
Forgot Password Login or registration. This is how the ‘scope trace looks when the potentiometer is turned all the way anti-clockwise to its limiting position. Turn the potentiometer clockwise “right” carefully, just a little at a time! Only registered users can write reviews. To test the frequency stability, a simple 30m band direct conversion receiver is used, powered from a well-regulated linear power supply.
AD MHz DDS Module
Upon switch on of a new AD module of course, there is some warm-up drift of the MHz crystal reference oscillator, which shows as drift of the ARGO line.
Be the first one to review this product. Guess you like Empty. All the way clockwise: Orders usually shipped in 1 business day! Oscilloscopes Affordable portable and bench scopes–DIY kits available! The following examples are NOT correct oscilloscope waveforms: This is how the ‘scope trace looks when the potentiometer is turned all the way clockwise to its limiting position.
Some kit builders have noted poor output on 10m, which has been resolved by very very slight adjustment of the comparator potentiometer. Coming Soon QSX all-band all-mode transceiver. The result should be a sharp, straight line without any instabilities or wiggles.
All the way anti-clockwise: Where To Buy Nooelec. Note that for these tests, the Ad8950 PA is actually disconnected.
The description below may help people who need to adjust an AD module for use in an Ultimate3 kit. Additional Info SKU The signal at the DDS squarewave output pin 7which is also the same as microprocessor input pin 11, is monitored on a MHz oscilloscope via a x10 probe.